Question regarding a FLuke A meter display. «on: January 28, , 04 am». So I’ve gotten into restoring “vintage” DMMs from Ebay. I’ve gotten. Download FLUKE A IM.V4 SERVICE MANUAL service manual & repair info for electronics experts. Hi guys. I managed to buy a Fluke A for £1, knowing that it wasn’t working. The selection switches were broken bar one, and they were.
|Genre:||Health and Food|
|Published (Last):||13 January 2018|
|PDF File Size:||5.25 Mb|
|ePub File Size:||14.35 Mb|
|Price:||Free* [*Free Regsitration Required]|
Bedford, Ohio Eimac Div. Williamantic, Connecticut Nytronics Inc. City of Industry, California Vernitron Corp. Connect shield of the input leads to the guard terminal. Total Quantity per assembly or component. Box 30, Concord, N.
Geneva, New York Ohmite Mfg. If reshipment becomes necessary, the instru- ment should be repackaged in the original container. Adjustment names given in this procedure agree with those silk-screened onto the top guard cover.
Contact your nearest Sales Representative for additional information.
The deluxe test lead kit, shown in Figurecontains two test leads with probes red and black and five pairs of universal probe tips. Table 7A-1 defines the assembly revision levels documented in this manual with an X. The current to be applied to the knowm and unknown resistance is supplied by a constant current source consisting of a voltage regulator, Q21 and Qland a current regulator, Q2 and CR2.
AC Line Connection DC Linearity Check Therefore, its trailing edge is recommended for use as a print command to the recorder. A malfunction in the A fouke be isolated to a particular section of circuitry by observing the displayed symptoms during the multimeter operation in each function.
You may also like.
Fluke A Angle Position Indicator w/On- Board Ref.,Progrmable FLV $ For Rent | Transcat
Adjust the kfl cal R8 for an A display of Box Wellington, New Zealand Tel: The range adjustment procedure ensures the ac measurement accuracy of each available range. Each option and accessory is listed by Model or Option number in the table of contents on page Manufacturer’s part Number or Type.
Apply a kfl standard resistance to the input. Short pins 3 and 4 of U17 together.
Fluke 8810A FOR RENT
Dallas, Texas Solitron Devices Inc. Maximum calibrator accuracy is attained by using the divider for lower voltage requirements. The probe tips include: Input Terminal Connections A The following procedure is used to gain access to the interior of the A: These older meters used to be built around a or chip which was known to blow easily mine’s on it’s third.
Adjust the 10 kfl cal R10 for an A display of Lid Joseph Zatzman Dr. Chicago, Illinois Optimax Inc. I’ve read the manual http: NOTE The 88 10 A front panel includes all of the controls, indicators, and connectors required to accommodate the maximum legal combination of options, i. Chicago, Illinois Centrelab Electronics Div. For more information on Fluke 881a or Sales Offices you may dial toll-free in most of U. Trending Price New. Theory of Operation Controls, Indicators, and Connectors S i Table See fouke 5 in any Fluke technical manual for ordering instructions.
This section contains information regarding the installation and operation of the Model A. The Ohms Converter determines the value of an unknown resistance by comparison. Compare the current DVM reading with that taken in step 3. If requested, an estimate will be provided to the customer before any repair work is flkue on instruments that are beyond the warranty period.
The guard terminal on the A front panel can be connected in a way that provides a signal path for the common mode current other than through the input leads carrying the normal mode voltage signal. The scope control settings are provided on the wave shape illustration.